The PowerPac™ HV power supply supports an output of 5,000 V, 500 mA, and 400 W, which allows its use for all high-voltage applications, including low-current applications in the microampere range. It is ideal for IEF and DNA sequencing. With 400 W output, the PowerPac HV offers enough power to run the most demanding IEF experiments or up to four DNA sequencing cells simultaneously.
Features and Benefits
Innovative, wireless run data transfer is possible with optional PowerPac data transfer software, which makes it easy to organize, display, print, analyze, export, and annotate run data from the PowerPac HV power supply. Easily transfer run data from anywhere in the laboratory using the power supply's built-in wireless IR capability. Data can be sent first to a PDA and then to a PC or directly to a PC with a peripheral IR receiving device.
PowerPac data transfer software allows you to:
- Examine and track the programmed method
- Examine actual run conditions (voltage, current, method changes, error conditions) recorded periodically during the run
- Annotate a run with user notes
- Identify the serial number of the unit that generated the data
- Document all run-related information with reports
- Organize your run data into projects
- Customize data representation
To support IQ/OQ within GLP- and FDA-regulated environments, the PowerPac HV IQ/OQ protocol and test box are available for use exclusively with the PowerPac HV power supply. The test box houses a factory-calibrated resistive load. When used with the appropriate meters to monitor output voltage, current, and power, the test box determines the power supply's performance against its specifications.
The optional temperature probe is used with the PowerPac HV power supply to monitor gel temperature between 30 and 90°C during electrophoresis. Temperature control is particularly useful for DNA sequencing and SSCP analysis, in which constant temperature is important for resolution and reproducibility.
Applications and Uses
- Nucleic acid gel electrophoresis
- Protein gel electrophoresis
To select the power supply that best fits your needs, see the Power Supply Selection Guide or the PowerPac Power Supply Comparison chart.