Proteins in a sample can be quantified using different assays. This section describes available assays for protein concentration determination including the classic Bradford and Lowry assays.
Related Topics: Cell Disruption, Protein Solubilization, Removal of Interfering Substances.
Determine the concentration of protein in a sample by protein assay to:
The most commonly used protein assays are colorimetric assays in which the presence of a protein stain causes a color change, proportional to the amount of protein in the sample, that can be measured with a spectrophotometer. All protein assays utilize a dilution series of a known protein (usually bovine serum albumin (BSA) or bovine γ-globulin), to create a standard curve from which the concentration of the sample is derived (see protocol tab). For additional information on protein quantitation assays see Sapan et al. 1999 and Noble and Bailey 2009.
The chemical components of the sample buffer and the amount of protein available for assay dictate the type of assay that may be used (see table and Selection Guide below).
*The Bradford assay is, however, highly sensitive to ionic detergents like SDS (see table below).
To measure protein concentration in Laemmli buffers, use the RC DC protein assay, which is compatible with reducing agents and detergents. For more information on protein quantitation using colorimetric assays, refer to Bio-Rad bulletin 1069.
Bio-Rad offers a number of protein assays based on the classic Bradford and Lowry assays. The choice of the assay will depend on a number of factors including the concentration of the proteins in the sample and the presence of reducing agents and detergents.
All assays are easy to use, require little reagent preparation, and give accurate and reproducible results. * To adapt the RC DC assay to a microplate format, follow the micro test tube (microfuge tube) assay protocol in the RC DC instruction manual up to the centrifugation step where the supernatant is discarded. The pellet can then be transferred to the microplate, and the microplate assay protocol in the DC protein assay manual can be followed. In a typical protein assay, a chemical reagent is added to a protein sample, producing a visible result, such as a color change in the sample solution. This color change is quantitated with a spectrophotometer or microplate reader, and compared to a standard curve of known concentrations of protein versus their absorbance after reaction with the reagent. The amount of protein in the unknown sample is determined by interpolation, reading the concentration of protein on the standard curve that corresponds to its absorbance.
* Nonidet P–40 >0.25%, Triton C-100 >0.05%, Tween 20 >0.01%, SDS >0.025% ** DTT >1 mM, EDTA >1 mM
The color change observed in protein assays is monitored and quantitated using a spectrophotometer. Spectrophotometers such as Bio-Rad's SmartSpec™ Plus Spectrophotometer may have preprogrammed methods for protein quantitation to facilitate data collection and analysis.
SmartSpec Plus spectrophotometer.
Berkelman T (2008). Quantitation of protein in samples prepared for 2-D electrophoresis. Methods Mol Biol 424, 43–49.
Bradford MM (1976). A rapid and sensitive method for the quantitation of microgram quantities of protein utilizing the principle of protein-dye binding. Anal Biochem 72, 248–254.
Noble JE and Bailey MJ (2009). Quantitation of protein. Methods Enzymol 463, 73–95.
Sapan CV et al. (1999). Colorimetric protein assay techniques. Biotechnol Appl Biochem 29, 99–108.
Smith PK et al. (1985). Measurement of protein using bicinchoninic acid. Anal Biochem 150, 76–85.
Call us at 1-800-268-0213
Please reenter your email address in the correct format.
Please enter your email address.
Your subscription information already exists, we will send you an email with specific instructions to manage your existing subscription profile.
To receive the latest news, promotions, and more, sign up for Bio-Rad updates by entering your email address below. You can elect to receive only the types of Bio-Rad communications that are of interest to you..
Copyright © 2015 Bio-Rad Laboratories, Inc. All rights reserved.